FRESENIUS 2008K MANUAL PDF
We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||13 July 2018|
|PDF File Size:||6.80 Mb|
|ePub File Size:||10.60 Mb|
|Price:||Free* [*Free Regsitration Required]|
Arterial Chamber This option is used to define whether the arterial chamber is pre-pump or post-pump. Sensor Calibration Procedures Add to My Favorites. Modules Chapter 1—Overview Modules The modules accompanying the K are located just below the control panel.
With some RO systems, the water inlet line may be disinfected along with the RO and distribution piping manusl leaving the dialysis machine in Rinse mode during RO disinfection.
K High Flow Error Possible balancing chamber problem. Use Table 27 in Appendix B to ensure that you have compatible concentrates and configurations. Flow Relief Pressure UF Removed is set to 0.
Page of Go. It takes approximately 15 years for a new device to become K approved by the FDA.
FRESENIUS MEDICAL CARE K OPERATOR’S MANUAL Pdf Download.
Be sure the end of the segment clears the bottom of the yoke. To shift the window downward, press the Down arrow key. Secure the patient end of the venous line over the priming bucket.
The K hemodialysis machine can be set up to perform online PHTs during treatment. The customer fresenuis the user of the K hemodialysis machine should assure that it is used in such an environment. Using cuff tubing longer than 10 feet may result in erroneous blood pressure readings. Section 15 – Blood Leak Problems Section 10 – Pressure Tests Failing Post Temperature Sensor Calibration Single needle lines usually have an 8 mm ID pump segment Perform all tasks necessary in setting up the K machine for dialysis, i.
Fresenius Medical Care 2008K Operator’s Manual
The water inlet line is part of the water mxnual system and is not disinfected by the dialysis machine. Replacing The 9-volt Battery Recommended Screen Order Page TP that is to be replaced freseniua each treatment. Be sure to open the door to the optical detector before pulling the line from the venous clamp and optical detector assembly. Section 6 — Concentrate Mnaual Errors I manaul odog is right.
Section 18a – Diagnostic Valve Leak Tests UF rate will be maintained even with low blood pump speeds. The UF Time button acts as a countdown timer displaying the remaining time ultrafiltration will be performed. Electronic Circuit Description Using A Central Venous Catheter The Control Panel In the event of a failure, an alarm sounds and a warning message is displayed in the status box. If unable to reset the alarm, call your local qualified technician.
The K can be configured in Service mode so that audible alarms occur only when the optical detector senses blood. In some cases, the operator must reset them.
During an override, the blood leak detector is inactive. Chapter 3—Setting Treatment Parameters New Treatment Key When the K is first turned on in preparation for daily operation after a long power downall treatment parameters revert to their default settings.
Got it, continue to print. It is the intention freeenius Fresenius USA to improve products continuously, a process which may result in modifications to specifications or equipment produced in the future.
Alarm 2 Check for a disconnected line. Hold maual one second to turn the power off and If blood is sensed, the machine will power down with an audible alarm. Temperature Control Calibration Single Needle Dialysis optional Don’t have an account?
The K hemodialysis machine is equipped with both visual cues and audible alarms to alert the operator to potential problems. Place the dialysate supply lines on the shunt 2. Product Improvement Policy To automatically turn the machine off after the Auto Heat Disinfect program runs, select the Off after Heat Disin option in Service Mode see figure When problems or potential problems are detected, the operator is alerted through informational messages displayed on the screen and in some cases, audible alarms.
If warning is not cleared, take the machine out of service and alert a qualified technician. It’s not Nanual thats preventing a new machine from hitting the market, it’s the FDA.
Organization Of The k About This Manual This manual is organized to systematically guide a patient-care specialist through the set up, operation, and clean up of the K in daily use.